Skip To Main Content
Frequently Asked Questions:
Main FAQ Page
How often should I calibrate the analyzer?
How much flow do I need to get a good sample?
What are the IR shim and CH
How do I set the device time on the analyzer?
How do I set up nitriding calculations on my analyzer?
Can my analyzer do dew point?
What is the recommended pressure for calibration gases?
How do I use the CO
scrubber for calibrations?
How do I create and start a session?
How to I connect to XGA viewer software?
When should I change the sample tube filter?
I upgraded to Windows 10 and now my device won't connect to XGA Viewer. What do I do?